SEARCH WITHIN CONTENT
Citation Information : Australasian Orthodontic Journal. Volume 31, Issue 2, Pages 165-170, DOI: https://doi.org/10.21307/aoj-2020-151
License : (CC BY 4.0)
Published Online: 15-August-2021
Aim: The aim of the present study was to characterise the chemical and mechanical properties of contemporary thermoplastic orthodontic materials.
Materials and methods: Four thermoplastic materials were tested: Clear Aligner (Scheu-Dental), ACE and A+ (Dentsply), and Invisalign (Align Technology). Eight appliances were fabricated from each material and a small portion from each was analysed by ATR-FTIR spectroscopy. The appliances were cut and, following metallographic grinding and polishing, were subjected to instrumented indentation testing (IIT) employing a Vickers indenter. Martens Hardness (HM), Indentation Modulus (EIT), Elastic to Total Work Ratio (elastic index (ηIT)) and Indentation Creep (CIT) were determined according to ISO 14577-1. The mean values of the mechanical properties were statistically analysed by one way ANOVA and Tukey Kramer multiple comparison test at a = 0.05.
Results: ATR-FTIR analysis identified that Invisalign was a polyurethane-based material, whereas the others were based on polyester, polyethylene glycol terephthalate (PETG). Invisalign showed higher hardness and modulus values, a slightly higher brittleness and lesser creep resistance compared with the PETG-based products.
Conclusions: The materials tested showed significant differences in their chemical structure and mechanical properties and therefore differences in their clinical behaviour are anticipated.
1. Bergström K, Halling A, Wilde B. Orthodontic care from the patients’ perspective: perceptions of 27-year-olds. Eur J Orthod 1998;20:319-29.
2. Ziuchkovski JP, Fields HW, Johnston WM, Lindsey DT. Assessment of perceived orthodontic appliance attractiveness. Am J Orthod Dentofacial Orthop 2008;133:S68-78.
3. Walton DK, Fields HW, Johnston WM, Rosenstiel SF, Firestone AR, Christensen JC. Orthodontic appliance preferences of children and adolescents. Am J Orthod Dentofacial Orthop 2010;138:698 e1-12; discussion 698-9.
4. Rosvall MD, Fields HW, Ziuchkovski J, Rosenstiel SF, Johnston WM. Attractiveness, acceptability, and value of orthodontic appliances. Am J Orthod Dentofacial Orthop 2009;135:276 e1-12; discussion 276-7.
5. Kravitz ND, Kusnoto B, BeGole E, Obrez A, Agran B. How well does Invisalign work? A prospective clinical study evaluating the efficacy of tooth movement with Invisalign. Am J Orthod Dentofacial Orthop 2009;135:27-35.
6. Simon M, Keilig L, Schwarze J, Jung BA, Bourauel C. Treatment outcome and efficacy of an aligner technique—regarding incisor torque, premolar derotation and molar distalization. BMC Oral Health 2014;14:68.
7. Vardimon AD, Robbins D, Brosh T. In-vivo von Mises strains during Invisalign treatment. Am J Orthod Dentofacial Orthop 2010;138:399-409.
8. Simon M, Keilig L, Schwarze J, Jung BA, Bourauel C. Forces and moments generated by removable thermoplastic aligners: incisor torque, premolar derotation, and molar distalization. Am J Orthod Dentofacial Orthop 2014;145:728-36.
9. Hahn W, Engelke B, Jung K, Dathe H, Fialka-Fricke J, KubeinMeesenburg D et al. Initial forces and moments delivered by removable thermoplastic appliances during rotation of an upper central incisor. Angle Orthod 2010;80:239-46.
10. Hahn W, Dathe H, Fialka-Fricke J, Fricke-Zech S, Zapf A, KubeinMeesenburg D et al. Influence of thermoplastic appliance thickness on the magnitude of force delivered to a maxillary central incisor during tipping. Am J Orthod Dentofacial Orthop 2009;136:12 e1- 7; discussion 12-3.
11. Kohda N, Iijima M, Muguruma T, Brantley WA, Ahluwalia KS, Mizoguchi I. Effects of mechanical properties of thermoplastic materials on the initial force of thermoplastic appliances. Angle Orthod 2013;83:476-83.
12. Clements KM, Bollen AM, Huang G, King G, Hujoel P, Ma T. Activation time and material stiffness of sequential removable orthodontic appliances. Part 2: Dental improvements. Am J Orthod Dentofacial Orthop 2003;124:502-8.
13. Fang D, Zhang N, Chen H, Bai Y. Dynamic stress relaxation of orthodontic thermoplastic materials in a simulated oral environment. Dent Mater J 2013;32:946-51.
14. International Organization for Standardization. Metallic materials – Instrumented indentation test for hardness and materials parameters. ISO14577-1. Geneva: ISO, 2002.
15. Gracco A, Mazzoli A, Favoni O, Conti C, Ferraris P, Tosi G et al. Short-term chemical and physical changes in invisalign appliances. Aust Orthod J 2009;25:34-40.
16. Gardner GD, Dunn WJ, Taloumis L. Wear comparison of thermoplastic materials used for orthodontic retainers. Am J Orthod Dentofacial Orthop 2003;124:294-7.
17. Ryokawa H, Miyazaki Y, Fujishima A, Miyazaki T, Maki K. The mechanical properties of dental thermoplastic materials in a simulated intraoral environment. Orthodontic Waves 2006;65:64-72.
18. Pascual AL, Beeman CS, Hicks EP, Bush HM, Mitchell RJ. The essential work of fracture of thermoplastic orthodontic retainer materials. Angle Orthod 2010;80:554-61.
19. Kwon JS, Lee YK, Lim BS, Lim YK. Force delivery properties of thermoplastic orthodontic materials. Am J Orthod Dentofacial Orthop 2008;133:228-34; quiz 328 e1.
20. Shahdad SA, McCabe JF, Bull S, Rusby S, Wassell RW. Hardness measured with traditional Vickers and Martens hardness methods. Dent Mater 2007;23:1079-85.