SEARCH WITHIN CONTENT
Citation Information : International Journal on Smart Sensing and Intelligent Systems. Volume 9, Issue 2, Pages 1,110-1,129, DOI: https://doi.org/10.21307/ijssis-2017-910
License : (CC BY-NC-ND 4.0)
Received Date : 15-January-2016 / Accepted: 20-March-2016 / Published Online: 01-June-2016
In order to realize the human body temperature fast and non-contact measurement, an infrared thermometer is designed. The infrared human body temperature sensor is mainly used to convert the human body's infrared into voltage signal, an operational amplifier to amplify the signal, filter circuit to filter the signal, the analog signal into digital signal by the A/D conversion circuit, data processing by the MCU, LCD display and voice reporting body temperature and time, so the human body non-contact measurement is realized. The experimental results show that: the device can realize the temperature and time of acquisition, the measurement error is not more than 0.5℃, voice broadcast and liquid crystal display the temperature and time, overrun alarm and other functions.
 C.L. Hao, Y.W Shen, “Study on Production and Mercury Consumption of Clinical Mercury-
Thermometers in China”, Research of Environmental Sciences , vol. 19,no. 1, 2006,pp.18-21.
 M. Yan, C.W. Peng, Y.H. Yan, etal ,“The principle of infrared temperature measurement and
error analysis”, Journal of Hunan University,vol.31,no.5, 2004,pp.110-112.
 Y. Cui, H.F. Li, “The principle of infrared temperature measuring instrument and Application
Guide”, Automation & Instrumentation , total 14,no.6,2009,pp.103-107.
 J. Li，M.D. Liu, Y.K. Zeng, etal, “Study on non contact infrared temperature measurement ”,
Piezo-electrics & Acoustooptics,vol.23,no.3,2001, pp.202-205.
 W. Chen, “The Intelligent Monitoring System for Piglets Based on Automatic Control
Algorithm”, Hangzhou Dianzi University，2012.
 W.P. Liu, “Measurement and Control Node of oil Automation System Design and
Implementation ”, He Fei University of Technology，2007.
 J.Z. Li, “Principle and Practice of Micro controller unit”, Publishing House of Xidian
University，Second Edition, 2008.
 W. Wang, S.R. Li, “Temperature acquisition and wireless transmission based on 8051 MCU ”,
Modern electronic technology,vol.34,no.1, 2011, pp. 146-149.
 B. Marco, B. John, O.F. Brendan, H. Nicholas and D. Paul , “Low-power wireless liquid
monitoring system using ultrasonic sensors”, International Journal on Smart Sensing and
Intelligent Systems, vol. 8, no. 1, March 2015,pp. 26-44.
 Z.H. Xin, G.L. Chen, H. Li, Q.X. Song, L.Y. Hu, etal, “The smart home system based on the
IAP15f2k61s2 and GSM”, International Journal on Smart Sensing and Intelligent Systems, vol. 7,
no. 4, December 2014,pp. 1789 – 1806.
 D.G. Yang, T.y. Liu , C.Y. Hu, “Design of an intelligent car for searching track and avoiding
obstacles”, International Journal on Smart Sensing and Intelligent Systems, vol. 8, no. 2, June
2015,pp. 1354 – 1378.
 A.J. Shan,“Application of SYN6288 Chinese Speech Synthesis Chip in Smart Sowing
Monitor”, Agricultural Science & Technology and Equipment, total 216,no.6,2012 ， pp.43-
 K.Y. Guo, X.J. Chen, B.Ji, “Design of voice broadcast system about car’s speed based on
SYN6288”,Experimental Technology and Management, vol.31,no.1,2014,pp.87-90.
 X.D. Zhuo, “Design of speech recognition system based on SPCE061A ”, Measurement
and testing technology, vol.39,no.3,2012，pp.5-8.
 B. Jiang, “Study and Implementation of the Intelligent control system of Antimony Oxide
Production Process”, Guangxi University，2008.
 D.J. Feng,“The Design of Electronic Clock Based on DS1302”,Science &technology
 C. Su,“Production of LCD display calendar based on the MCU controlling ”, Journal of
Yangtze University (natural science edition), vol.10, no. 22, 2013, pp.77-79
 L.Z. Rao, “Design of a Multifunction Infrared Phonetic Thermometer ”, Journal of Sanming
University, vol.23, no. 2, 2006, pp.148-152.
 C. Chen. “Study on Protel DXP Packaging Components Library”, Modern Electronic
Technique, vol. 32, no. 24, 2009，pp.163-167．
 W.X. Cao, “Design of digital thermometer based on AT89C51 MCU ”, Journal of Wuhan
Engineering Institute, vol. 24, no. 3, 2012, pp.32-34.