SEARCH WITHIN CONTENT
VOLUME 7 , ISSUE 5 (December 2014) > List of articles
Special issue ICST 2014
Citation Information : International Journal on Smart Sensing and Intelligent Systems. Volume 7, Issue 5, Pages 1-5, DOI: https://doi.org/10.21307/ijssis-2019-070
License : (CC BY-NC-ND 4.0)
Published Online: 15-February-2020
A dual slope direct digital converter (DSDDC) suitable for resistive sensor elements already connected in a bridge form is presented. A conventional dual slope analog to digital converter (DS-ADC) is altered with an intentionally introduced instrumentation amplifier (INA) to realize the proposed DDC. The DSDDC accepts bridge connected resistive sensor elements as input and provides a digital output that is proportional to the quantity being sensed by the sensor elements. The results obtained from simulation of the proposed DSDDC indicate a worst case error in the output to be < ±0.03 %. Experimental results from a prototype unit, presented herein, demonstrate the practicality the scheme. Worst case error of the prototype unit was found to be < ±0.07 % with test conducted using standard decade resistance boxes emulating the sensor elements. The worst case error in the prototype was < ±0.05 % when tested with a load-cell made of four strain gauges.
 E. O. Doebelin, Measurement Systems—Application and Design, 5th ed. New York: McGraw-Hill, 2004.
 B. Maundy and S. J. G. Gift, “ Strain Gauge amplifier circuits,” IEEE Trans.Instrum. Meas. September 2013, vol. 62, no. 4,pp. 693-700.
 S. Gift and B. Maundy, “ New configurations for the measurement of small resistance chaneges,” IEEE Trans. Circuits Syst. II Analog Digit.Signal Process.,vol. 53,no. 3pp. 178-82, Mar 2006.
 D. Stefanescu, “Strain gauges and Wheatstone bridges basic instrumentation and new applications for electrical measurement of non – electrical quatities,” in Proc. 8th Int. Multi-Conf. SSD,Mar. 2011, pp. 15.
 Walt Kester , Practical design techniques for sensor signal conditioning, Analog devices, 1999.
 Agilent Technologies, “Practical strain gauge measurements”, Application Note 290-1, 1999.
 E. W. Owen, “An integrating analog – to digital converter for differential transducers,” IEEE Trans. Instrum. Meas.,vol. IM-28, no. 3,pp. 216-220, Sep. 1979.
 Watanabe, Kenzo; Mochizuki, Kouji, “ A high resolution, linear resistance to frquency converter,” IEEE Trans.Instrum.Meas., vol. 45(3), 1996, pp. 761-764.
 S. Kaliyugavaradhan , “ A linear resistanc to time converter with high resolution,” IEEE Trans. Instrum. Meas., vol. 49, #1, Feb. 2000, pp. 151-153.
 R. P. Areny and J. G. Webster, “Digital and intelligent sensors,” in Sensor and Signal Conditioning (2nd ed. ). Toronto, Ontario, Canada: Wiley, 2001, pp. 458–478.
 S. Yurish, “Universal smart sensor interface and signal conditioner,” in Proc. IEEE Sensors Conf., Atlanta, GA, USA, 2007, pp. 24–27.
 Hewlet Packard, “Fundamentals of time interval measurements,” ApplicationNote 200-3, Jun. 1997.
 N. M. Mohan and V. J. Kumar, “Direct digital converter for a single, active element resistive sensor,” in Proc. IEEE I2MTC, Singapore, 2009, pp. 828–831.
 N. M. Mohan, B. George and V. J. Kumar, “A novel dual slope resistance to digital converter,” IEEE Trans. Instrum. Meas., vol. 59,no. 5,pp. 1013-1018, May 2010.
 P. R. Nagarajan, B. George and V. J. Kumar, “ A direct digital converter for resistive sensor elements in bridge configuration,” in Proc. IEEE ICST, Wellington , Newzealand, 2013, pp. 516-519.
 ATmega328 Data Sheet, Atmel Corporation – http://www.atmel.com/images/doc8161.pdf
 Michael Margolis and Nicholas Weldin, “Arduino Cookbook”, (1st ed. ) O‟Reilly Media, March 2001.