SEARCH WITHIN CONTENT
VOLUME 7 , ISSUE 5 (December 2014) > List of articles
Special issue ICST 2014
Citation Information : International Journal on Smart Sensing and Intelligent Systems. Volume 7, Issue 5, Pages 1-5, DOI: https://doi.org/10.21307/ijssis-2019-083
License : (CC BY-NC-ND 4.0)
Published Online: 15-February-2020
In this paper, a MEMS based capacitive nasal sensor system for measuring Respiration Rate (RR) of human being is developed. At first two identical diaphragm based MEMS capacitive nasal sensors are designed and virtually fabricated. A proposed schematic of the system consists of signal conditioning circuitry alongwith the sensors is described here. In order to measure the respiration rate the sensors are mounted below Right Nostril (RN) and Left Nostril (LN), in such a way that the nasal airflow during inspiration and expiration impinge on the sensor diaphragms. Due to nasal airflow, the designed square diaphragm of the sensor is being deflected and thus induces a corresponding change in the original capacitance value. This change in capacitance value is to be detected by a correlateddouble-sampling (CDS) capacitance-to-voltage converter is designed for a precision interface with a MEMS capacitive pressure sensor, followed by an amplifier and a differential cyclic ADC to digitize the pressure information. The designed MEMS based capacitive nasal sensors is capable of identifying normal RR (18.5±1.5 bpm) of human being. The design of sensors and its characteristics analysis are performed in a FEA/BEA based virtual simulation platform.
 Milic-Emili J,“Recent advances in clinical assessment of control of breathing”, Lung 1982, pp. 160:1 – 17.
 D. Moffett, S. Moffett and C. Schauf, “Human physiology”, 2nd edition, Wm. C. Brown Publishers, 1993, Ch. 13 and 16.
 Ivana Prkic et al.,”Pontine ȝ-opioid receptors mediate bradypnea caused by intravenous remifentanil infusions at clinically relevant concentrations in dogs,” J Neurophysiol 2012, vol.108, pp. 2430 –2441.
 Weigt SS, Abrazado M,Kleerup EC, Tashkin DP, Cooper CB,” Time course and degree of hyperinflation with metronomepaced tachypnea in COPD patients,” COPD 2008, vol. Oct;5(5),pp.298-304. doi: 10.1080/15412550802363428.
 Folke M, Cernerud L, Ekstrom M, Hok B,” Critical review of non-invasive respiratory monitoring in medical care,” Med Biol Eng Comput 2003,vol. 41,pp.377-383.
 R. Farre , J.M. Montserrat, D. Navajas,”Noninvasive monitoring of respiratory mechanics during sleep,” Eur Respir J 2004,vol. 24,pp. 1052–1060.
 H.Kawai, “The piezoelectricity of Polyvinylidene fluoride”, J.appl.phys. 1969,vol.,8,pp.73-77.
 Siivola J,”New non-invasive piezoelectric transducer for recording of respiration, heart rate and body movements. Med Bio Eng Comp 1989,vol. 27,pp.423-424.
 Choi S, Jiang Z,:”A novel wearable sensor device with conductive fabric and PVDF film for monitoring cardiorespiratory signals,” Sensors and Actuators A: Physical 2006,vol. 128,pp.317-326.
 J.H. Oum, S.E. Lee, D.-W. Kim and S. Hong;” Non-contact heartbeat and respiration detector using capacitive sensor with Colpitts oscillator”, ELECTRONICS LETTERS 17th January 2008 Vol. 44 No. 2.
 Anurekha Sharma,”Modelling of a Pull-in voltage and touch point pressure for MEMS capacitive transduces with square diaphragm”, Kurukshetra University, Kurukshetra,. Ph.D. thesis July 2008.
 Roopa et al., “Idetification of different respiratory rate by a piezo polymer based nasal sensor”, IEEExplore 978-1-46734642-9/13.